Glass Slide Arrayers
V&P Scientific is proud to introduce the first economical and practical way to make cDNA, DNA, RNA or protein microarrays on glass slides. Whether you want to make large arrays on glass slides, work out new protocols for your robotic Glass Slide Arrayer or just Q/C new DNA preparations, this simple and flexible system allows you to tackle both large and small jobs without major setup hassles.
|Convenience||Versatility||Enhanced Features||Spot Size||Spot Pitch||Accessories Available||Custom Options|
|Print your own microarrays anytime from 96 or 384 well plates||Can spot your samples as duplicates, triplicates or more||Print up to 768 spots per array. Print duplicate slides at the same time.||~500 microns in diameter on silanated slides and 780 microns on Epoxy coated slides||1,125 microns on horizontal axis and 750 microns on the vertical axis, can change pitch
easily for different spot sizes by skipping rows or columns
|Indexing System for 96 and 384 well source plates to simplify arraying and enhance spot size reproducibility. Replicator pin wash station to enhance arraying speed.||
Can be adapted to your robotic system, change pins for different applications, e.g., larger volumes
Our floating Pin Replicator can be fxitured with either 2 rows of 4 pins (8 total on 9mm centers for 96 well microplates (VP 478)
or 4 rows of 8 pins (32 total) on 4.5mm center for 384 well microplates (VP 478A). Then used in conjunction with the unique
V&P Scientific Glass Slide Indexing System VP 470 (US Patent #6,613,284 and Republic of China Invention Patent # NI-164,388,
and European patent #1,363,737) this dual slide system places 768 spots ~500 microns in diamter, onto a grid 18mm wide and
36mm tall on two separate slides.
The glass slide to the right illustrates the 768 spotting pattern on a silanated glass slide made by Xenopore
The glass slide to the right is an epoxy coated slide from Xenopore spotted with 5mm 24mer oligonucleotide
with a 3' amine linker
and a 5' Cy5 label in 50% DMSO.
The GenePix 3 software was used to determine a mean spot diamter of 780 microns with a CV of 5.5%.
|This is an enlarged area of the photo above.|
Aldehyde (Silanated) Slide with Amino/Cy5-Labeled Oligo
768 spot array, 5' Cy5 and 3' Amino linker
Fluorescence Data from GSI
Lumonics Scan of the slide
Accuracy and Precision Data from the Slide above Spot Diameter